# Difference between revisions of "Apogee Alta Test Report"

From Telescope

Line 1: | Line 1: | ||

<table> | <table> | ||

− | <tr><td width="100"><b>Initial Temp</b></td><td width="100"><b>End Temp</b></td><td width="100"><b>Time (min)</b></td></tr> | + | <tr><td width="100"><b>Initial Temp</b></td><td width="100"><b>End Temp</b></td><td width="100"><b>Time to cool (min)</b></td></tr> |

<tr><td>22.7</td><td>0</td><td>49</td></tr> | <tr><td>22.7</td><td>0</td><td>49</td></tr> | ||

<tr><td>0</td><td>-5</td><td>36</td></tr> | <tr><td>0</td><td>-5</td><td>36</td></tr> | ||

Line 15: | Line 15: | ||

<br> | <br> | ||

+ | All the temperatures are in Celsius. | ||

The read noise is the standard deviation of the master bias image (9 combined images). The dark current is the mean of the bias-subtracted, median-combined darks times the gain (1.3) divided by the integration time (300 seconds). The dark current is in electrons per pixel per second, and the read noise is in ADU r.m.s. All of these measurements are below the values given in the camera test report. | The read noise is the standard deviation of the master bias image (9 combined images). The dark current is the mean of the bias-subtracted, median-combined darks times the gain (1.3) divided by the integration time (300 seconds). The dark current is in electrons per pixel per second, and the read noise is in ADU r.m.s. All of these measurements are below the values given in the camera test report. |

## Revision as of 09:21, 3 August 2009

Initial Temp | End Temp | Time to cool (min) |

22.7 | 0 | 49 |

0 | -5 | 36 |

-5 | -10 | 39 |

Temperature | Dark Current | Read Noise |

0 | 0.078 | 6.935 |

-5 | 0.041 | 6.706 |

-10 | 0.035 | 6.631 |

All the temperatures are in Celsius. The read noise is the standard deviation of the master bias image (9 combined images). The dark current is the mean of the bias-subtracted, median-combined darks times the gain (1.3) divided by the integration time (300 seconds). The dark current is in electrons per pixel per second, and the read noise is in ADU r.m.s. All of these measurements are below the values given in the camera test report.